AmonMed Self Testing Kits Sperm Concentration Test Kits CE Gold Rapid Test Kit antigen rapid test
|Categories||Epidemic prevention materials|
|Name||TNN kit rapid test hbsag hiv hcv/with high quality|
|Brand Name||TNN or customized logo|
|Certificates||CE mark / FSC / ISO 13485/ CMDCAS|
|Test Sample||whole blood/serum/plasma|
|Shelf Time||24 --36 month|
|Available Format||Single strip,single dip-card,cassette,multi panel|
|Using place||home use or hospital/clinics|
EVANCARE kit rapid test hbsag hiv hcv/with high quality
HBsAg test is a rapid direct binding test for the visual detection of hepatitis B antigen (HBsAg) as an aid in the diagnosis of hepatitis B infection.It is based on the principle of sandwich immunossay for determination of HBsAg in human serum specimen.Two monoclonal antibodies are employed to identify HBsAg specifically.This one step test is very sensitive and only takes about 15-20 minutes.
TNN kit rapid test hbsag hiv hcv/with high quality
TNN or customized logo
CE mark / FSC / ISO 13485/ CMDCAS
24 --36 month
Single strip,single dip-card,cassette,multi panel cassette
home use or hospital/clinics
How to use it ?
For HBSAG test :
Allow the pouched strip/cassette to equilibrate to room temperature (2-30℃) prior to testing.before testing.
Collect whole blood/serum/plasma into a collection tube (containing EDTA,citrate or heparin) by venipuncture.
If specimens are not immediately tested,they should be refrigerated at 2~8℃.For storage periods greater than three days,freezing is recommended.They should be brought to room temperature prior to use. Using the specimen after long-term storage of more than three days can cause non-specific reaction.
When stored at 2~8℃,the whole blood sample should be used within three days.
Directions for Use:
1. Open a pouch containing a cassette, remove the test kit from the pouch and place it horizontally on the desk.
2. Pipette one drop(5µl) of specimen into the same well and then add one drop(50µl) of diluent into the same well.
3. Read results within 15 minutes. Do not read results after 20 minutes.
Interpretation of Results:
Negative: Only control pink band appears on test region of the cassette. This indicates that there is no determinand in the specimen.
Positive: Two pink bands(C,T) appear on test region of the cassette. This indicates that the specimen contains detectable amount of determinand.
Invalid: If without colored band appears at control region, this is an indication of a possible error in performing the test. The test should be repeated using a new one.